2. Product Name and Specifications
2.1. Name and model
Name: Skyray Energy Dispersive X-Ray Fluorescence Spectrometer
2.4. Technical Performance and Specifications
2.4.1. The elements can be analyzed from the sodium (Na) to uranium (U)
2.4.2. The element content range from 2ppm to 99.99% can be analyzed.
2.4.3. Measurement time: 5-300s
Alloy: Mg base alloy, Al base alloy, Fe base alloy, Ni base alloy, Cu base alloy, Zn base alloy, Co base alloy, Zr base alloy, W base alloy, Ti base alloy, Precious Metal, rare earth metal and etc.
Mineral, Soil: Fe mine, Ni mine, Au mine, Cu mine, Zn mine, Ta/Nb mine and etc.
Simplified Spectrometer structure
3.1 Silicon Drift Detector + DMCA analyzer
Detector converts x-rays into electrical pulses. DMCA process the signal from detector and send to microprocessor for analysis.
3.1.1. Electrical Peltier cooling detector; Resolution: 145±5eV
3.1.2 DMCA: Digital multi-channel analyzer
3.2 X-Ray Tube
3.2.1. Lifetime is longer than 20,000 working hours.
3.2.2. 50 watts beryllium window with Rhodium target X-ray tube. (Silver, tungsten target is optional)
3.3 HV supplier (Spellman USA)
3.3.1. Provide 0-50kV power to X-ray tube to create primary x-ray.
3.3.2. Long term stability<0.01%
3.4 High Resolution CCD Camera
3.4.1 Observing sample.
3.5.1 Eight collimators (Diameter of 0.1, 0.5, 1, 2, 3, 4, 6, 8mm) can be switched automatically by software for different application
3.6.1 Five different filters can be switched automatically by software.
3.7 Free Combinations between Collimators and Filters
3.7.1 There are 40 combinations of the collimators and the filters.